General Features and Graphical User Interface
In addition to the MINI-PAM-II, the WinControl-3 software operates the MICRO-PAM, MONITORING-PAM, DIVING-PAM-II and JUNIOR-PAM fluorometers, PAM-CONTROL operated instruments (MICROSCOPY-PAM, MICROFIBER-PAM and WATER-PAM Fiber Version) as well as the Universal Light Meter ULM-500.
Saturation pulse analysis and automated experimental protocols can be performed under control of the software WinControl-3 but also by the MINI-PAM-II internal software which is active when the MINI-PAM-II is operated autonomously. In the latter case, experimental parameters are entered using the transflective touchscreen. Continuous recording of fluorescence and modeling of light curves, however, requires WinControl-3.
Saturating pulse analysis with automatic detection and calculation of standard fluorescence parameters: F0, FM, F0’ (measured or calculated), FM’, F, FV/FM, qP, qL, qN, NPQ, Y(II), Y(NPQ), Y(NO), ETR.
Export in CSV or TXT format of original fluorescence traces, saturating pulse analysis data and parameter estimates of light response curves.
Repetitive triggering of many fluorometer functions (e.g., saturation pulse analysis, induction curves). Automatic execution of light and induction curves, offset correction and calibration of internal light sensor.
Customer-defined Measuring Protocols
Execution of customized experimental procedures using batch files.