MONITORING-PAM AQUATIC Version
The ONLINE configuration of the AQUATIC version of the MONITORING-PAM permits measurement of photosynthesis of submersed plants. The ONLINE configuration requires external power supply and is operated by a Windows computer. The interface MONI-IB4/LANS permits connection to a Windows computer via Ethernet, RS-232 or USB communication.
Up to four waterproof emitter-detector heads MONI-HEAD/S can be linked to the interface MONI-IB4/LANS. For this purpose, the interface provides four input sockets to connect underwater cables RS-485/S. The water depth of measuring sites is determined by the length of the RS-485/S underwater cables which is 10 m by default.
MONI-HEAD/S Measuring Head of the AQUATIC versions.
MONI-HEAD/S Measuring Head of the AQUATIC versions. Close-up with titanium sample clip.
The STAND-ALONE configuration of the AQUATIC version of the MONITORING-PAM is suited for long-term monitoring of chlorophyll fluorescence at water depths down to 75 m. The system consists of the waterproof battery-powered data acquisition system MONI-DA/S, the PC interface MONI-IB1/S, and up to seven waterproof emitter-detector heads MONI-HEAD/S.
The MONI-DA/S runs for more than 4 weeks on battery with seven MONI-HEAD/S connected and a saturation pulse interval of 15 minutes. Data are logged on an internal ring buffer and on a non-removable microSD card. Data are transferred from the MONI-DA/S to a computer via the interface MONI-IB1/S. Unlike the MONI-DA of the TERRESTRIAL version, wireless data transfer is not feasible underwater.
Back view of MONI-DA/S.